The ozone therapy device OZON 2000 is manufactured by Zotzmann + Stahl GmbH & Co. KG, Winnenden.
It is the first hyperbaric ozone therapy device with type approval according to MedGV.
The safety concept of the OZON 2000 excludes damage to the patient.
The therapy device OZON 2000 is characterized by the easiest operation via a wipeable membrane keypad.
OZON 2000 is not an automatic treatment machine with a confusing program flow. The doctor himself controls the course of each treatment step – this is the greatest security for the patient.
|Radio protection test||Radio protection sign of the VDE (Association of German Electrical Engineers)|
|Device type||OZON 2000|
|Voltage||220 V, 50 Hz|
|Output||max. 120 VA|
|Fuse||2 x 0,63 A – slow|
|Protective measure||Protection classI|
|Degree of protection||covered version|
|Concentration range||0 – 70 µg/ml O3/O2|
|Gas flow||approx. 55 l/h|
|Weight||approx. 13 kg|
|Feed gas||only medical oxygen|
|Pressure of the feed gas||fixed by special pressure reducer|
|Technical subject to change without prior notice.|
The complete equipment of the OZON 2000 includes:
The technical support is provided by Bernd Hese Medizintechnik, Bückeburg.
The quality assurance of the Ozon 2000 is carried out by constantly VDE-monitored production. OZON 2000 is not an automatic treatment machine with a confusing program flow – the doctor himself controls the course of each treatment step, which is the greatest security for the patient. The device monitors itself and the therapist during treatment. Damage to the patient is excluded.
During therapy, the doctor has to press the footswitch every 8 seconds to confirm his presence. If no switching pulse occurs, the treatment is automatically stopped after another 8 seconds.
The AIR-BLOC system additionally prevents gas from flowing in the direction of the vein.